Portable Outlet Oxygen Concentration Detector Oxygen Monitor In Fish farm O2
|Usage||To test the oxygen output of oxygen generator|
|Power off automatically||in 15mins without single|
1, Adopt 3 years imported long-life electrochemical oxygen sensors.
2,This tester needs to correct before use, please check the correction instruction;
3, Please replace the battery if it's low, or the numerical value will show incorrectly;
4, This device will auto shut down after turning on for 30 mins.
1, Turn on the device, connect the pipette filler with the tester and pinch for 2-3 times to suck the fresh air; 3 mins later, if the numerical value is not within 20.9±0.5, then the tester needs to correct.
2, Turn on the tester, simultaneously press "Key 1" and "Key 2" for 5 seconds, then loosen the keys and enters the correction function.
3, Correction: press "Key 1" to increase 0.1%, press "Key 2" to decrease 0.1%, until correct to 20.9+-0.05, then shortly press the left key to exit. Then it's in test condition.
Our Service :
1.Reply your inquiry in 24 working hours.
2.Experienced staffs answer all your question in fluent English.
3.Customized design is avaliable.OEM&ODM are welcomed.
4.Exclusive and unique solution can be provided to our customer by our well-trained and professional engineers and staffs.
5.Special discount and protection of sales area provided to our distributor.